Please use this identifier to cite or link to this item:
|Title:||Raman Laser Spectrometer (RLS) calibration target design to allow onboard combined science between the RLS and MicrOmega instruments on the ExoMars rover|
|Authors:||López Reyes, G.|
Moral, A. G.
Manrique, J. A.
Bibring, J. P.
Rodríguez, J. A.
Pérez Canora, C.
Mateo Martí, Eva
Prieto Ballesteros, O.
Vago, J. L.
|Keywords:||Calibration terget;RLS;MicrOmega;ExoMars combined science|
|Publisher:||Wiley Analytical Science|
|Citation:||Journal of Raman Spectroscopy 51(9): 1718-1730(2020)|
|Abstract:||The ExoMars rover, scheduled to be launched in 2020, will be equipped with a novel and diverse payload. It will also include a drill to collect subsurface samples (from 0‐ to 2‐m depth) and deliver them to the rover analytical laboratory, where it will be possible to perform combined science between instruments. For the first time, the exact same sample target areas will be investigated using complementary analytical methods—infrared spectrometry, Raman spectrometry, and laser desorption mass spectrometry—to establish mineralogical and organic chemistry composition. Fundamental for implementing this cooperative science strategy is the Raman Laser Spectrometer (RLS) calibration target (CT). The RLS CT features a polyethylene terephthalate disk used for RLS calibration and verification of the instrument during the mission. In addition, special patterns have been recorded on the RLS CT disk that the other instruments can detect and employ to determine their relative position. In this manner, the RLS CT ensures the spatial correlation between the three analytical laboratory instruments: MicrOmega, RLS, and MOMA. The RLS CT has been subjected to a series of tests to qualify it for space utilization and to characterize its behavior during the mission. The results from the joint work performed by the RLS and MicrOmega instrument teams confirm the feasibility of the “combined science” approach envisioned for ExoMars rover operations, whose science return is optimized when complementing the RLS and MicrOmega joint analysis with the autonomous RLS operation.|
|Appears in Collections:||(CAB) Artículos|
Files in This Item:
|acceso-restringido.pdf||221,73 kB||Adobe PDF|
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.